Differentiation  Kind  Specification 
Procedure  Form group  10 groups 
The form section is counted  100 sections / 100 at most for each group Period of run time: It takes 99 hours and 59 minutes at most 
PID number  Temperature. The humidity is 6 groups respectively 
The form is connected  Each form group is joinable 
Input  Input signal Test the range   Input the kind  Range 
Temperature  Pt1100 Pt2100 DCV  50.00~150.00 100.0~200.0 1.000~5.000V ( 100.0~200.0 ) 
Humidity  Pt1100 DCV  0.0~100.0 (0.0~100.0% RH) 1~5V(0.0~100.0%RH) 
Control accuracy  Temperature: 0.1%ofFS Humidity: 1%ofFS 
Control cycle  500ms 
Output  Control and export  2 o'clock (temperature, the humidity) 
Output kind  Outcoming signal (SSR) : ON voltage =Above 15VDC, minimum Pulse width: 5ms 
Convey the specification  Convey and export  2 o'clock (temperature, the humidity) Output the form: PV,SP,MV 
Outcoming signal  4~20mADC 
Load resistance  Maximum 600 
The trouble is input  Input the kind (can presume the failure message name)  1.RUN/STOP 2 DI2 (error information 2) 
3 DI3 (error information 3) 4 DI4 (error information 4) 
Enter mode  The capacity of joint: Maximum 12VDC, 10mA 
The joint exports  The joint exports  The maximum of joint is at 12 o'clock (Relay: At 8 o'clock, OpenCollector: 4 o'clock) Festival capacity: Maximum 30VDC/1A, 250VAC/1A 
Output kind (amount to 30 o'clock)  1.Alarm: 4 o'clock (can presume alarm type)  
2 IS (internal joint) : 6 o'clock 
3 TS (sequence joint) : 2 o'clock 
4.Temperature Run: 1 o'clock 
5.Humidity Run: 1 o'clock 
6.Temperature Up: 1 o'clock 
7.Temperature Down: 1 o'clock 
8.Humidity Up: 1 o'clock 
9.Humidity Down: 1 o'clock 
10.Temperature Soak: 1 o'clock 
11.Humidity Soak: 1 o'clock 
12.Drain: 1 o'clock (can set for time) 
13.Trouble: 1 o'clock (can set for time) 
14.patternEnd: 1 o'clock (can set for time) 
15.1stRef: 1 o'clock (can set for time) 
16.2ndRef: 1 o'clock (can set for time) 
17.ON/OFF: The temperature is at 4 o'clock, the humidity is at 1 o'clock (can set for time) 
Power  Power  Nominal voltage: 100~250VAC ( 10%) ,50/60Hz, Consume the electricity: Maximum 15W 
Lithium cell  Presume Data keeps and uses (at least 10 years) 
Insulation resistance  500VDC20M 
Withstand voltage  2000VAC/1 minute 
Communication  Class of traffic  RS485 (Line 2) The maximum 1.2Km of communication distance, joinable 31 at most, Maximum 9600bps of communication speed 
Type  Option  Explain in detail 
TEMI300  00  Standard 
TAG:TEMI300
